lrs2551 create module-name ?options...?
lrs2551 config module-name ?options...?
lrs2551 cget module-name
The lrs2551 command allows you to create and configure LeCroy LRS 2551 scaler modules. The LRS 2551 is a 12 channel scaler. While it is normally read in the periodic scaler stack, it is also possible to read it in the event stack when properly configured.
The create subcommand creates a new module
assigning it the name
module-name should be used to refer
to the module in future commands. Optional configuration data
can also be appended to the create subcommand.
The config subcommand configures an existing
module-name. The configuration items are
name value pairs. See OPTIONS below for more information about
the options supported by the lrs2551 command.
The cget subcommand returns the current module configuration. The configuration is returned as a Tcl list. Each element of the list is a two element sublist that contains the configuration name and value in that order.
The lrs2551 command supports the following configuration options:
-cumulative is false (default),
the scaler module is cleared after each readout providing
the counts between each readout as the value of each
channel. This shoulid be false when used with the NSCLDAQ
scaler display program in the scaler stack.
Supplies the virtual slot number; vsn for the module. This only needs to be supplied if the module is going to be read in the event stack. Note that currently unpacking scaler data from the event stack is not supported.
If true (the default is false),
the virtual slot number set by the
configuration value is inserted in the data block from
the scaler. If false it is not.
If used with the NSCLDAQ Scaler display in the scaler stack,
this should be false.
When installed in the CAMAC crate, the module should
be inserted in the
by this option.
This example creates an LRS 2551 scaler: