A fundamental limitation placed on the VM-USB is that only one process can
communicate with it at a given time. This introduces a complication when
trying to use separate programs for controlling devices that need only to
be configured rather than read out. There many ways to deal with this
issue, but the approach taken by VMUSBReadout is to turn the VMUSBReadout
program into a proxy for other applications that might want to use the
VM-USB. In another sense, the VMUSBReadout serves the VM-USB to other
programs. This is accomplished via the slow-controls server that listens
and accepts connections from clients who make remote requests to it. The
port on which the server listens defaults to 27000 but can be defined by
--port command line argument to
If the run is halted, the slow-controls server can directly manipulate the VM-USB to perform the desired device changes. If, however, the VMUSB is in data taking mode, the slow-controls server pauses the run, performs the requested operation, and then resumes the run. You should avoid working with control panels during production runs for that reason.
The configuration of the slow-controls subsystem is similar to that of the readout subsystem. A configuration file, written in Tcl, defines the control modules present in the system. An important difference to the daqconfig.tcl file is that the controls configuration file is read only once when VMUSBReadout starts. For that reason, modules loaded into the slow-controls server are configured once at start up and do not change during the duration of VMUSBReadout.
The Module command is used to create, configure, and query the configuration of slow-controls devices known to the server. A set of prepackaged module types are provided by VMUSBReadout. Each created module must be given a unique name and is automatically registered to the server upon creation. When a client makes a request to the slow-controls server, it must provide the name of the client it wishes to handle its request. If no module with the name the client provides has been registered to the server, unexpected results are likely to occur as well as an error. For that reason, if you wish to use the slow-controls server, it is important that you register the module with the server first before using your client. Client requests either get, set, or update requests.
Up to this point, we have mostly discussed how the slow-controls server acts as a proxy for other client applications. However, the slow-controls subsystem also has some utility in its own right. For one, the watch command can be used to periodically poll devices for status without pausing data taking. This is useful for monitoring the performance of devices over the coarse of data taking. The subsystem is also ideal for performing tasks that need to occur only once during a session or on demand. A good example of this is the loading of firmware. The vast majority of times, firmware only needs to be loaded once per power cycle. For this reason and the fact that it often takes a noticeable amount of time to complete, it makes little sense to do it every run. The slow-controls server is a good tool for this kind of operation because slow-controls modules each define an initialization procedure that is executed at start up or anytime thereafter via the init command.
Once again, it is simplest to illustrate how to set up the controls configuration script by example rather than in prose. For that reason we will proceed with a short example.
Example 4-2. A Sample Controls Configuration
In this example, we will set up the server to configure a Caen V812 constant-fraction discriminator. The Caen V812 is a VME board that responds to commands addressed to it. The board that we are configuring has a base address of 0x12450000, which we will need to provide to the module we create. The type of module that understands how to work with the V812 has a module type of "caenv812" so we must pass that type as an argument to the Module command. To create the module and configure its from a file we add the following two lines to our ctlconfig.tcl script.
Module create caenv812 cfd Module config cfd -base 0x1245000 \ -file [file join [file dirname [info script]] cfd1.cofig]
That is it. The act of creating the module also registers it by the name cfd to the slow-controls server. At this point, a client application could in principle attempt to send it requests.
If a user wants to write their own module for the slow-controls subsystem, they can do so. There is a well-defined framework for developing these. You can learn more about creating your own modules at Extending the slow controls subsystem.
There are numerous slow-controls modules availabe for use with the Module command (see Module(3tcl)). Because of its usefulness, we will very briefly mention the "vmusb" module type. This module enables a client of the slow-controls server to execute arbitrary command stacks as remote procedure calls. This functionality must be enabled by including the following line in your control configuration file:
Module create vmusb ctlr
You can see that there is no magic here compared to the previous example. We have just created a new module named "ctlr" of type vmusb. This module type is the handler for the remote procedure calls and literally only knows how to execute lists and return the results. Clients that want to submit their remote requests then do so by using the following classes:
Instances of this class represent lists of VME operations you want to perform via the server. See CVMUSBReadoutList for reference information.
Instances of this class represent connections to VMUSBReadout slow
controls servers. Instances provide a mechanism for submitting a
CVMUSBReaoutlist for remote execution and
gathering the results back for client code.
See CVMUSBRemote for reference information about this class.
Programs using these classes must link to the libVMUSBRemote library in the NSCLDAQ lib subdirectory.